IOGP SAFETY ALERT
OVERPRESSURE OF AN EXPLOSION-PROOF ENCLOSURE
|Incident Date: 15 May 2017 Time: —|
|Type of Activity: —|
|Type of Injury: Pressure release|
|An analyzer engineer was fatality injured when the engineer removed the cover on an explosion-proof enclosure as part of a routine task. The threaded cover, weighing ~12 lbs. and measuring 14 inches in diameter, was forcefully propelled from the enclosure as the engineer unscrewed it, inflicting a fatal head injury. The force was caused by pressure inside the enclosure from leaking sample gas or instrument air components. The enclosure was not equipped with an external indicator to determine the pressure inside the enclosure and had no means to relieve internal pressure.|
Corrective Actions and Recommendations:
Figure 1: Re-enactment of cover removal
Figure 2: Example of Explosion-Proof Enclosure
safety alert number: 288
With a wealth of experience in the oil and gas industry, we provide independent materials and corrosion consultancy to help companies reduce risk, save money and apply best practice in their oil and gas businesses.
Find out more by getting in touch – you’ll get an answer from a qualified, experienced materials engineer every time.
How to get in touch
US: 877-399-1010 (toll free)
66 Eldon St